Atrato Ultrasonic Flow Meter Electrical Installation (Page 3)
This page references Atrato ultrasonic flow meter electrical installation. The Atrato ultrasonic flow meter can be used entirely from a computer using the USB connection but this may not be acceptable in many situations where remote operation or further functions are required.
The unit will work with systems from Windows XP onwards. The maximum connector cable size is 2.5mm (22-14AWG), for ease of assembly we recommend 1mm maximum. During electrical installation, care should be taken when terminating the wires as the conductors should be stripped to 4 to 5mm maximum and the wire ends must be pushed fully into the connector before tightening. These cables pass through a slot cut in the aluminium housing please ensure that no bare conductors are clear of the surface of the connector strip prior to closing the housing.
Operates on External Power OR USB
Operates on External Power Only
View the complete Titan Flow Meters Range
Next page: Ultrasonic flow meter setup
Previous page: Ultrasonic flow meter installation